Measurement Tension Amplifier Work With Tension LoadCell for
Measuring amplifiers ensure a stable bridge supply voltage, They
amplify the output signal of the load sensors to a standard signal
Tension transducer STA - 05 is the tension of Intelli gent tension
transducer instrument . By using of key operation will complete
zero alignment,calibration,convert the left or right tension sensor
signal into astan dard industrial control signals, 0 ~ 10V / 0 ~ 5V
and 0 ~ 20 mA output signal can be provided to PLC for automatic
1. 32 bit CPU guarantees high speed, precision, ant-ijamming
capability and ensures reliability.
2. The outside has no adjustable potentiometer.
3. The tension sensor can select Differential sensor or Strain-
4. 0~ 10V/ 0~ 5V and 0~ 20mA output signals can be p-rovided to PLC
and display instrument.
5. Use dispensing technology, small volume, light wei-ght.
6. Using DIN rail mounting, simple and convenient.
|Power Supply||DC 24V|
|Input Analog singal||L+~L-;R+~R-;Left,right STS tension loadcell signal (0~200mv,power
supply DC5V)||L+~L-;R+~R-;Left,right STSA tension loadcell signal (0~15mv;power
|Fixed output||5V+-GND:DC5V tension loadcell power supply|
|2||24V - : Negative terminal of tension transducer po -|
|3||Mc1 : Select output voltage range .|
|MC1 with MCC disconnect : output 0 ~ 10V.|
|MC1 with MCC connect : output 0 ~ 5V.|
|4||Mc2 : Select the calibration value .|
|MC2 with MCC disconnect : Full scale ca -|
|MC2 with MCC connect : Half scale calibr -|
|5||Vout : Positive||terminal||of||output||voltage , 0 ~ 10V or|
|0 ~ 5V.|
|6||GND : N egative||terminal||of||output||voltage / current .|
|7||Iout : Positive||terminal||of||output||current , 0 ~ 20mA .|
If STA - 05 tension transducer install STS Differe - ntial sensor
by wuhan true engine company , wir - ing colors as follows :
“5V +”— red ,“0V” — black ,“Terminal +”— yellow ,“Terminal -”—
If STA - 05 tension transducer install STSA Strai - ngauge sensor
by wuhan true engine company , wiring colors as follows :
“5V +” —red , “0V” — black , “Terminal +”— green , “Terminal -”
If only use one tension sensor , please use wires to connect L +
with R + , L - with R -
Notes of Using Tension Detector
1. Set screws A and B should not be too long when the bearing
houses are installed on the detector. Other-wisethe force could not
cause relevant deformations and displacements on the detector. Use
spherical u-niversal bearing houses to avoid the detectors on b-oth
sides interacting on each other, causing errors on calibration.
2. The wrap angle of film over detector reel shouldn’t be larger
than 120, around 60°would be perfect. The s-ignal detected by
detector would be too weak to con-trol tension effectively with a
large wrap angle.
3. The load force caused by film tension should not poi-nt to the
screw ends of tension detector. Otherwise t-here will be very low
or even no output.
4. The screw ends of tension detector should point upw-ards when
installed on a wall.
5. If the film is pulled cross the reel from below, the fo-rce
caused by film tension pulls but not pushes the tension detector.
So the negative and positive detec-tor signal wires should exchange
when connected to the controller terminals.
Packing and delivery:
1. Neutral carton packing
2. One piece in one carton box
3. Normally one controller with two loadcells,this is easy for
4. One unit shipping
5. Lead time:within 3 days